Agilent technologies HP B Pdf User Manuals. View online or download Agilent technologies HP B Operating Manual. HP B Synthesizer Function Generator. Service Manual. Up for sale is the following Technical Manual in original binder/binding Original Manufacturer’s. I recently won a “for parts or repair” HP B function generator for a great Although free scans of the B service manual are readily.
|Published (Last):||9 September 2011|
|PDF File Size:||20.67 Mb|
|ePub File Size:||12.36 Mb|
|Price:||Free* [*Free Regsitration Required]|
Basic Troubleshooting Procedure 8-A-l. Amplitude Modulator Any equipment that satisfies the critical specifications given in the table may be substituted servife the recommended model. A26 Assembly SA1 Signatures.
HEWLETT PACKARD B INSTALLATION SYNTHESIZER/FUNCTION GENERATOR SERVICE MANUAL | eBay
Measure voltage level at A3U18 pin 9. Amplitude Flatness Writes data to A21 U1 325b, executes a sweep, and reads data back. Test Equipment for Troubleshooting X Drive Equipment Required: It multiplies the waveform by the amplitude value that U latched on to. Non-Listed Parts 6- 8. To change a part on the Keyboard assembly, first remove the Display assembly, then remove the 13 screws holding the assembly to the front ho.
The width of the positive pulse should decrease to zero, then reset and repeat at a 1 Hz rate TTL levels. No front panel display or annunciatiors. Voltage at A3U 1 8 pin 6 should be low, pin 9 should be low.
Because of the increased complexity of the logic circuits used to control many instruments, malfunctions in these circuits may be very difficult to locate. Observe the spectrum analyzer for spurious responses. Learn More – opens in a new window or tab Any international shipping is paid in part to Pitney Bowes Inc. Test Equipment for Troubleshooting.
Agilent Technologies 3325B Synthesizer/function Generator
In normal operation, the sigiial at the collector of Q should be essentially the same as the signal at the junction of R and R These voltage levels are useful when troubleshooting amplitude problems. Because of its small size, it is often damaged when being soldered. If the voltage is still incorrect, the fault lies prior to TP6. Repeat the ramp jitter measurements of Steps d 2 uid f, adjusting RAMP ADJ as necessary to reduce the jitter to 60p,s or for the best compromise between the two.
Press the Local key to exit Special Test Mode If the level is incorrect, troubleshoot the API 5 sub-block and the U19 programming signals. Also make sure the microcircuit packages that are mounted in sockets are firmly seated.
If the signal is good, proceed to step u. Power Supplies Service Group O Program an instrument sweep from IkHz to 20MHz while monitoring the signal at the center servuce of J Learn More – opens in a new window or tab. If the waveform is correct, the amplifier is probably operating correctly, and the problem may be in the Attenuator, Service Group L.
Refer to eBay Return policy for more details. Our service centers may be able to perform calibration if no repair parts are needed, but no other support from Agilent is available.
HP B Synthesizer Function Generator Service Manual P/N | eBay
Set the main signal to 10 MHz. The number on the right side of the display should read between 0. Wervice data is now in U, and the register address is in U Frame or chassis terminal.
Before starting a Signature Analysis test, use the following steps to help you sservice the problem. Table 6r3 lists parts in alphanumeric order of their reference designators and indicates the description, -hp- part number of each part, together with any applicable notes, and provides the following: Select a valid country.
During continuous sweep, the ramp is reset at the start of sweep down.